This manual provides documentation for the following instruments: EA (3 Hz – GHz). EA (3 Hz – GHz). EA (3 Hz – GHz). EA (3. This manual provides documentation for the following instruments: PSA Series. EA (3 Hz – GHz). EA (3 Hz – GHz). EA (3 Hz – GHz). This manual provides documentation for the following instruments: EA (3 Hz – GHz). EA (3 Hz – GHz). EA (3 Hz – GHz). EA (3.
|Published (Last):||12 June 2005|
|PDF File Size:||20.55 Mb|
|ePub File Size:||1.59 Mb|
|Price:||Free* [*Free Regsitration Required]|
If a problem persists, you may choose to: Accessories Sales and Service Office and are listed below. When you have completed the printer setup for the analyzer, press the front panel key.
AGILENT TECHNOLOGIES EA GETTING STARTED MANUAL Pdf Download.
Copying a File 1. Agilent Technologies makes no warranty of any kind e444a0 regard to this material, including but not limited to, the implied warranties of merchantability and fitness for a particular purpose. You e4440a now press the NOTE You need to use a filename that does not already exist in the current directory. Contacting Agilent Technologies Online assistance: Page 53 NOTE You should notice that the trace you saved, in this example trace 3, is loaded to trace 2.
For a complete list of spectrum analyzer messages, see the Instrument Messages and Functional Tests manual.
Used Agilent EA
Line voltage does not need to be selected. Initial Inspection Initial Inspection Inspect the shipping container and the cushioning material for signs of stress. Page of 80 Go. To see that trace 2 is in view mode, blank traces 1 and 3 by pressing.
Turning on the Analyzer for the First Time Plug in the power cord. This function can be auto-coupled. Options and Accessories Ordering Options and Accessories Ordering Options and Accessories Options and accessories help you configure the analyzer for your specific applications, and enable you to use the analyzer. Check to see if the green LED above the power switch is on. Directories are displayed as Manula and drives have nothing listed under Type:.
To access the Copy menu, press boxes as described by the From: Protecting Against Electrostatic Discharge Protecting Against Electrostatic Discharge Electrostatic discharge ESD can damage or destroy electronic components the possibility of unseen damage caused by ESD is present whenever components are transported, stored, or used.
In this example you will save a trace to the internal C: Using The Front Panel Using the Front Panel Entering Data When setting measurement parameters, there are several ways to enter or modify the value of the active function: Place a marker labeled 1 on the 10 MHz peak, as shown in Press Peak Search Note that the frequency and amplitude of the marker appear both in the active function block, and in the upper-right corner of the screen.
The only green key is the key, which performs an analyzer reset A summary of all front Preset Page 18 When you have completed the printer setup for the analyzer, press the front panel key. Viewing A Signal Viewing a Signal 1.
WARNING If this product is to be energized via an external auto transformer for voltage reduction, make sure that its common terminal is connected to a neutral earthed pole of the power supply. Each softkey More has several characters on it.
Creating A Directory b.
This analyzer does not contain customer serviceable fuses. Page 34 Making a Basic Measurement This chapter is divided into the following sections: Firmware Revision To view the firmware revision of your analyzer, press call Agilent Technologies regarding your analyzer, it is helpful to have this revision and the analyzer serial number available. Installation and Setup Running Internal Alignments Running Internal Alignments Each time the analyzer is powered on, the internal alignment routine runs automatically.
While janual all of the zeros following the decimal in the active function block are significant, the numbers after the decimal in the marker annotation upper-right corner of screen are significant.
Agilent Technologies E4440A Getting Started Manual
In Case of Difficulty Before calling Agilent Technologies, however or returning an analyzer for serviceperform the quick checks listed in the problem. Agilent Technologies shall not be liable for errors contained herein or for incidental or consequential damages in connection with the furnishing, performance, or use of this material.
The amnual is located on the CPU board.