The ELx continues our tradition of offering the tightest specifications of any microplate reader. Visit BioTek to learn more about our precision microplate. ELx Operator’s Manual. Revision History. Rev Date Changes. A. 1/96 First Release. B. 5/96 Revised reader specifications. Added Scanning. View and Download Biotek ELx operator’s manual online. Absorbance Microplate Reader. ELx Laboratory Equipment pdf manual download. Also for.
|Published (Last):||18 October 2007|
|PDF File Size:||11.17 Mb|
|ePub File Size:||6.97 Mb|
|Price:||Free* [*Free Regsitration Required]|
Reinstall the fuse holder with the fuse toward the back of the power input module.
EQUIV if the sample is greater than or equal to 1. Define The Test Plate Parameters Curve Fit Report Replace the cover, and verify that the white plastic indicator peg lines up with the correct voltage indicator hole.
Robotic Motor Adjustment Procedure Assay Main Menu, and then select. Table of contents All Rights Reserved Robotic models should have certain components lubricated. Number Of Kinetic Points Selection Discard the used gloves and cloths, using a Biohazard trash bag and an approved Biohazard container.
The software will automatically increment each subsequent sample identification number by 1.
Microplate Readers: ELx Absorbance Microplate Reader – Overview
Dual wavelength readings can significantly reduce optical interference caused by scratched or fingerprinted microplates since the scratches or fingerprints reduce the amount of light on both wavelengths.
Some readers have custom programmed assays installed.
Another Transformation Example Reuse Of Standard Curves Waste Electrical And Electronic Equipment Instructions for Programming a New Assay Turn off the reader and allow the lamp to cool for at els808 15 minutes before attempting to replace it.
Table of Contents Add to my manuals Add.
Changes made to the information in this document will be incorporated in new editions of manuap publication. Meas Selection Valid ranges are: If there are any mismatches, correct them and re-run the test.
Perform The Tests Formula Type Chapter 3: To define a cutoff formula for Positive and Negative manuql Solution B Required Materials: Errors and indicate that a particular axis has moved to a point. Error Codes general Beginning The Plate Read Add 95 mL of DI water; cap and shake well. The performance characteristics of the data reduction software have not been established with any laboratory diagnostic assay.
BIOTEK ELX OPERATOR’S MANUAL Pdf Download.
Page Older-Style Module A failed fuse is usually an indication of another problem that a new fuse is not likely to remedy. Optional Connect Maanual 6: Lamp may be out or blocked by a faulty filter wheel.
Incubator Self Test Current Average: ELx Absorbance Microplate Reader Instrument Qualification Chapter 4 Instrument Qualification This chapter discusses the tasks and procedures necessary for qualifying instrument performance initially and on an ongoing basis. Empty Carrier Test